1. Accuracy
In this context, it refers to how close a position solution delivered by the softwaredefined GNSS receiver is to the true (actual) position. Hence, it is a measure of the bias or systematic error. Its measurement requires a reference (also known as fiducial) position in the case of static positioning, and a controlled mobile platform in the case of dynamic positioning.
The definition of the reference point implies the agreement on some reference coordinate systems for the satellite system and the reference position:
 GNSS satellite coordinate reference system: The International Earth Rotation and Reference Sytems Service (IERS) recommend to express it as “ITRFyy at epoch yyyy.y”^{1}
 A local geographic coordinate reference system (providing transformation parameters, if applicable).
 An EastNorthUp (ENU) reference frame with origin in the reference point.
 In case of differential GNSS configurations, datum of the differential source.
Upon those definitions, most common position accuracy metrics for 2D and 3D positioning, expressed in a local ENU reference frame, are defined below:
Measure  Formula  Confidence region probability  Definition 

2DRMS  95 %  Twice the DRMS of the horizontal position errors, defining the radius of circle centered at the true position, containing the horizontal position estimate with probability of 95 %.  
DRMS  65 %  The square root of the average of the squared horizontal position errors, defining the radius of circle centered at the true position, containing the horizontal position estimate with probability of 65 %.  
CEP  , accurate if  50 %  The radius of circle centered at the true position, containing the horizontal position estimate with probability of 50 %. 
99 % Spherical Accuracy Standard  99 %  The radius of sphere centered at the true position, containing the position estimate in 3D with probability of 99 %  
90 % Spherical Accuracy Standard  90 %  The radius of sphere centered at the true position, containing the position estimate in 3D with probability of 90 %  
MRSE  61 %  The radius of sphere centered at the true position, containing the position estimate in 3D with probability of 61 %  
SEP  50 %  The radius of sphere centered at the true position, containing the position estimate in 3D with probability of 50 % 
with the standard deviations, in the case of a static receiver, computed as:
where is the East coordinate of the reference location. Similar expressions can be defined for the North and Up coordinates:
Ideally, static position measurements should be averaged over 24 hours and performed on a clearsky environment.
In the case of a dynamic receiver, position measurements and references will have a time index. In order to mitigate differences due to satellite visibility and geometry, averaging realizations of the same trajectory and location, performed at different time schedules, is encouraged.
For collections of position measurements delivered at instants , and performed over repetitions of the same trajectory at the same pace:
where is the East reference position of trajectory at time , and is the position measurement in the East coordinate for trajectory collected at time . Similar expressions can be defined for the North and Up coordinates:
Indicators of Accuracy
Upon the definition of:
 The GNSS satellite coordinate reference system (expressed as “ITRFyy at epoch yyyy.y”^{1}) and ellipsoid (e.g., WGS 84);
 The local geographic coordinate reference system (providing transformation parameters, if applicable) and ellipsoid;
 In the case of static accuracy, a reference point, adequately surveyed and expressed in a well–defined coordinate reference system, and in which the Antenna Reference Point (ARP) is placed when collecting raw signal data, or as defined by the synthetic generator tool;
 In the case of dynamic accuracy, one or more reference timed trajectories, in which the Antenna Reference Point (ARP) was mechanically moved when collecting raw signal data, or as defined by the synthetic generator tool;
 In case of differential GNSS configurations, the datum of the differential source;
it follows a list of possible accuracy indicators for a softwaredefined GNSS receiver:
 Standalone static position accuracy.
 Position accuracy results are given in meters of error with respect to a reference (fiducial) point previously measured in a geodetic survey, or defined by the testing equipment. The most commonly used confidence measurements for 2D positioning are the Distance Root Mean Square (DRMS) and the Circular Error Probability (CEP); and the Mean Radial Spherical Error (MRSE), the Spherical Error Probable (SEP), and the 90 % and 99% Spherical Accuracy Standards when measures are expressed in 3D.
 Standalone dynamic position accuracy.
 In this case, the reference is not a single point but a timed trajectory. Different trajectories and time schedules can be averaged to mitigate differences due to satellite visibility and geometry. Same metrics than for static positioning, where the position references might now have time and trajectory indeces.
 Nominal mechanical accuracy of the controlled mobile platform moving the receiver while collecting measurements.
 Aided / DGNSS static position accuracy.
 Same metrics than in standalone static scenarios.
 Aided / DGNSS dynamic position accuracy.
 Same metrics than in standalone dynamic scenarios.
 Nominal mechanical and time accuracies of the controlled mobile platform moving the antenna while collecting measurements.
References

G. Petit and B. Luzum, Eds., IERS Conventions (2010), Verlag des Bundesamts für Kartographie und Geodäsie, Frankfurt, Germany, 2010, IERS Technical Note 36. ISBN 3898889896. ↩ ↩^{2}
Leave a Comment